To measure radiation it must be absorbed. The absorption is higher as more dens the absorber material is. There are gas filled ionisation chambers or Geiger tubes available but the most sensitive is a crystal of NaJ with a density of 3.7 g/cm3. This crystal produces a small light flashes when exposed to radiation which must be detected by a light amplifier called photo-multiplier. The whole detector must be build in a mechanical protection and light tight housing and also stabelized against any changes.
The two sources of radiation have to be shielded in lead containers to prevent the rays to travell in all directions. By a thin channel it is ensured that only in one direction the working beam is
allowed to come out. The radiation is measured by a scintillation counter which is the most sensitive detector available for industrial applications. Both, source and detector, is called measuring path. Two measuring
pathes are necessary for ash determination. It would be possible to use only one detector for both radiations but because physically in the detector each radiations influences the other it is more reliable to use
Either source or detector can be installed above or underneeth the belt. The measuring pathes should have a distance of 1 m minimum to each other. A firm mechnical frame is needed to prevent changes of the distance between source and detector: It must be absolutelly constant to avoid measuring errors.
©1999-2009 RGI GmbH Beermiß 15 D-75323 Bad Wildbad
Tel.: 49 7081 5026 Fax: 49 7081 5526
Webhosting and design by RPF Netdesign